Have a question? Call us or contact us here.
10 LPM Concentrator:
When you first get your system, it has been in uninsulated storage and transportation facilities. It can take up to 24hrs to warm up, and until then it may not function properly.
Proper functioning is that when you first turn on the machine, the lights will flash and go out. After a period of time that can be 1/2 minute to 5 minutes, the O2 light will come on. It may be accompanied by periodic beeps. This can continue for a while, 5 – 15 minutes. After that it should go off and stay off.
If the light is staying on after 15 or 20 minutes, one possibility is that there is moisture in the sieve beds. Try running it for 12 hours, turn it off for 8 – 12 hours, then turn it back on and see what happens.
If the flow ball has been set above the 10
Turn it down to 5. Push the reset button. Let it run for 15 minutes. If the yellow light goes out, turn it back up to 10, but not above 10.
20 LMP Concentrator
The 20 LMP configuration is made of two 10 LMP concentrator that
5 LPM Concentrator
Please call us if your 5 LMP concentrator has a yellow light to troubleshoot.